- Software reads "Waiting for Injection" after an injection is made and the injector is switched from Load to Inject
- 1515 Isocratic HPLC Pump
- Breeze 2
- Rheodyne manual inject valve
Inject start cable is not connected properly.
FIX or WORKAROUND
Ensure that the inject start I/O connection on the back of the 1515 pump (slots 1 and 2) are connected to the inject start I/O connection on the back of the detector