- ACQUITY UPLC Binary Solvent Manager (BSM)
- ACQUITY UPLC I-Class
- ACQUITY UPLC H-Class
Event 4 is always on (or closed) for all external valve positions. Hard wire the valve connection that should be connected to event 4 directly to ground. This will mimic the event being on all the time, which is required for all valve positions.
- SSV Event Switch Connections on an I-Class BSM. Note that the rear connection ports may differ between different solvent manager variants.
- The Solvent Selection Valve may also be installed on the Sample Manager.
- The Solvent Selection Valve must be configured in the method. See the table below for configuration schemes.
UPBINARY, UPBSM+, UPIBSM, UPIBSM, UPIBSM+, UPIBSM+, UPISMFL, UPISMFL+, UPISMFTIVD, UPISMFTN, UPISMFTN+, UPPBSM, UPQSM, UPQSM+, UPQSMBIO, UPQSMBIO+, UPSMFTN, UPSMFTN+, UPSMFTNB+, UPSMFTNBIO