- Instrument Failure when trying to run 2475 in Breeze
- Battery recently replaced and serial number reset in detector
- Breeze 2
Node configuration made when the 2475 did not have a serial number specified on the detector.
FIX or WORKAROUND
- Delete system in Breeze.
- Recreate system in Breeze using the 2475 with the serial number specified.
- Delete any instance of a 2475 serial number uninitialized.