- Alliance HPLC System
- 2695 Separations Module
The injection range for a standard loop (100 µL) is 0.1 to 100 µL.
- The volume of the syringe plays a role in the size of the injection volume per full stroke.
- If you want to do example, 500ul injections you can put a larger loop in place and use that for small and large injections. Leave the smaller syringe in place, and it will just draw a few times to get the correct amount for the injection volume