- Alliance HPLC System
- 2695 Separations Module
Alliance syringes have a single draw of up to 60% of the syringe volume.
Examples: 25-µLsyringe = 15-µL maximum draw 250-µL syringe = 150-µL maximum draw 2500-µL syringe = 1500-µL maximum draw
- With a 25-µL syringe and a standard loop, the syringe draws as many times as is necessary to fulfill the injection volume.
- When making a 75-µL injection using a 25-µL syringe and a 100-µL loop, five draws are made to equal 75 µL .
Syringe draw rate (µL/sec)
The draw rate of the syringe accommodates viscous samples. The rate for each selection adjusts to the size of the installed syringe.
Selections: Fast, Normal, and Slow