- Ghost peaks are present in blank injections with a 717 Autosampler
- The analyte dissolves in acetonitrile
- 717 Autosampler
There is contamination in the needle of the 717 Autosampler.
FIX or WORKAROUND
- Fill a vial with 100% acetonitrile.
- Perform 10 replicate injections to clean out the needle.