- "A failure occurred in requesting the instrument status" message appears in Empower 3 message center
- Instrument failure in Empower
- Run stops
- Empower 3
- Alliance 2695 separations module
Bad Ethernet connection between PC and 2695
FIX or WORKAROUND
- Reseat the Ethernet cable and restart the run.