- ACQUITY UPLC H-Class
- Sample Manager Flow-Through-Needle SM-FTN
- 100-µL extension loop
No, the syringe must be two times larger than the installed loop.
The 100-µL extension loop requires the 250-µL syringe, 410001347
For the various loop sizes, see KB 6746 What are the part numbers for H-Class Sample Manager Flow Through Needle extension loops