- 2424 lamp recently changed
- Peaks now show very high LSUs
- Peaks appear to be off the chart/flat-topped
- 2424 Evaporative Light Scattering Detector (ELSD)
The gain is set too high for the new lamp.
FIX or WORKAROUND
- Run the Auto Gain feature on the detector.
To run Auto Gain, press DIAG 3, and then press Enter. See additional notes in the operator's guide (page 95; 3-31).