- Instrument stuck in Source Standby
- Can control API and Collision gases
- Cannot switch to either Operate or Full Standby
- Electronics restart does not resolve
- UNIFI 1.8
- Xevo G2 XS Tof (but could conceivably affect other MS systems)
FIX or WORKAROUND
- Take system offline (in Device Management/Instrument Systems tab).
- In the Device Management/Workstation section, right-click the workstation device and choose Reset Device.
- Bring system online.
- You can now switch MS to Operate or Full Standby.