- Cannot put the MS into Operate
- MS error for status
- On the Diagnostics for the Lens Control PCB, there is a red LED for Interlocks indicating a microswitch error
- If the MS is connected to MassLynx, no error exists and the instrument can go into Operate
- Mass Spectrometer control
LND is locked up (even if it is a workstation)
FIX or WORKAROUND
- Go to Device Management and reset the workstation or LND if it is a work group environment.
- Put the instrument into Operate.