This article is part of a troubleshooting path. If the symptoms below do not precisely match your issue, or if the solution has not fixed the issue, consider starting at the beginning of the path using this link
- Frequent server busy errors
- Frequent communication losses with different ACQUITY modules
- Many losses on all components in NetEvaluator
- System loses its IP address
- ACQUITY system and mass spectrometer
- No internet connection; no antivirus software
Faulty network card
FIX or WORKAROUND
- Replace the network card.
See KCS article #1018, What is the NetEvaluator utility?