- MassLynx stops, and no further data is acquired
- Must reboot EPC before the system works again
- MassLynx 4.2 SCN 977
- Xevo TQ-S micro
Bug in SCN 977 software: an issue with certain dwell times causes a timing conflict.
FIX or WORKAROUND
- Fixed in SCN 1001 (TGR 2) although not recorded in the release notes.
- Workaround - Do not use the following dwell times:
0.041 s, 0.169 s, 0.233 s, 0.338 s, 0.402 s, 0.658 s , and 0.858 s
Dwell times from 1 ms to 1 s were tested; only the times noted above caused an issue.