- Cannot integrate the peak in the MassLynx Chromatogram window.
- Selecting the integrate button or going to Process > Integrate does not result in any peaks integration.
- MassLynx 4.1
The peak-to-peak amplitude is too high for the chromatographic peaks.
FIX or WORKAROUND
- In the MassLynx Chromatogram window, select Process and Integrate.
- Select Automatic noise measurement.
- Select Integrate.