- Inlet method Ready and OK lights are green
- System Status Ready light is green
- Activate the collection system
- Sample list pauses; can click Pause, but immediately pauses again
- No error messages
- MassLynx 4.1; no MS
- Waters Fraction Collector (WFC3 or 2767)
WFC3 bed is hung up.
FIX or WORKAROUND
Reset the beds:
- Open FractionLynx.
- Click Collection Control.
- Select Control > Reset Beds.
If emergency collection is used, you will need to reset the beds.
This article is also applicable to the 2767.