- When the console is opened, the following error message appears: "ACQUITY Console Client has Stopped Working"
- If the inlet is configured as a single-pump system (that is, not using Waters Pump Control), everything works properly.
- MassLynx inlet software produces no error messages.
- ACQUITY 2D System with Waters Driver Pack 2017 R1
Software defect in the ACQUITY Driver Pack 2017 R1. (This problem can occur on any MassLynx 4.1 or 4.2 SCN; there is no specific MS dependency for this issue.)
FIX or WORKAROUND
If updating the driver pack is not an option, consider downgrading the instrument control software version to Driver Pack 4 SR1.