- After saving an inlet method and reopening it, all of the details regarding temperature, column selection, and injectors are missing
- MassLynx 4.1 SCN 932
- SYNAPT G2-Si
- APGC source fitted
Corrupt MassLynx project
FIX or WORKAROUND
- Go to the main MassLynx page.
- Go to the File Menu.
- Select Project Wizard.
- Click Yes in the pop-up window called Select Project: When changing to a new Project services are automatically closed down. Continue?
- Specify a project name, description, and location for the project, and then click Next.
- Choose the option "Create using existing project as template" and browse for C:\MassLynx\Default.pro.
- Create a new method in the inlet method editor and click Save.
- Close the inlet method editor.
- Close MassLynx.
- Open the software and use as normal.