OBJECTIVE or GOAL
Calculate signal-to-noise (S:N).
- Open the chromatogram window for the run.
- Select Process.
- Select Signal to noise.
- Click in the signal window, and then right-click and drag across the peak.
- Click in the noise window, and then right-click and drag across a good 0.5 minute of noise.
- Select the processing parameters desired and select OK.
MLYNX, MLYNXV41, SUPMM