OBJECTIVE or GOAL
Configure inlet event logging to capture inlet errors for troubleshooting.
- Windows 7
MassLynx 4.1 has an advanced inlet logging utility that automatically runs in the default conditions.
Switch to advanced settings; run the configuration tool C:\MassLynx\InletEventLogConfiguration.exe.
Logging Levels: Set to Trace.
Logging Type: Set to Information.
Logging Time: Set to 4 or 5 days to preserve logs from weekend runs.
Disable any subsystems options that are not applicable (e.g., Solvent monitor Exe, Solvent Monitor Mgr, Solvent Monitor Kernel, CTC options if a CTC is not in use).
Click Apply, click OK, and then restart MassLynx.
In the event of an inlet error, collect the inlet log from C:\MassLynx\InletLogs (typical log name format is Log_YYYY_MM_DD.log).
- Each log file is collected for a 24-hour period.
- If there are no log files present in the InletLogs folder, it usually means that the logging levels are still set to the default, and that there have been no inlet errors.
- See the attached Inlet event logging.pdf for reference.