- In MassLynx > Chromatogram > Display > Analog... > Select Pressure trace
- Open or zoom in on the pressure trace
- Chromatogram window crashes with "not responding" message
- Must close and reopen MassLynx to recover
- Same data file crashes with MassLynx on a different PC
- A file collected on a different PC does not crash on this PC
- MassLynx SCN 905
- ACQUITY Pressure trace
MassLynx is acquiring corrupt data files.
FIX or WORKAROUND
- Reinstall MassLynx.