- Instrument failure
- Pressure failure
- Failure after 12 sample set lines run (including conditioning lines)
- De Vinci Beckman CE Control for Empower v1.2.1 driver
The cause of this defect is unknown.
It is related to the increment function and multi-line sample sets.
FIX or WORKAROUND
- Confirm that the driver was installed using Typical settings and not Custom settings. This may rectify the issue.
- Limit the total number of lines in the sample set to 12 or fewer, including conditioning lines.
- Ensure that each line is a single injection or action.
- Alternatively, inject one sample per sample set.