- In the instrument method, programmable events are set properly
- Neither Condition Column nor Equilibrate is executing the shutdown method to turn off the lamp
- Empower 3
- 996/2996 Photodiode Array (PDA)
In the General tab of the Instrument Method, the "Enable Events" check box is not selected.
FIX or WORKAROUND
- Select the "Enable Events" check box in the Instrument Method for the 996/2996 PDA.