- Problem extracting a derived channel beyond limits of 3D chromatogram error in message center when processing data
- No results generated in Empower
- Enough free project table space
- Derived channel is set appropriately at 210 nm
- Empower 2
- Empower 3
- Wavelength range is set from 200-400 nm in the Instrument Method.
- This range is too small for extracting a derived channel at 210 nm.
FIX or WORKAROUND
- Edit Empower Instrument Method and collect range from 190-400 nm.