- Unable to recover buffered data on LAC/E
- Manual instrument server recover command does not push data to the server
- Can ping server by LAC/E and LAC/E by server successfully
- Watersservice is running on theLAC/E
- Not all files are 1 kb in size
- Issue with only one LAC/E
- In the recover log it states "failed to recover sample set and unable to store vial"
- Empower 3 Enterprise
Low project tablespace for the project that contained the files that were buffering on the LAC/E
FIX or WORKAROUND
- Increase the project tablespace. Files should recover and be pushed up to the server after 10 minutes.