- Diverter valve goes to the correct position at start of method
- Diverter valve changes to the designated position per the Event Table
- Diverter valve does not change to the designated position per the last event in the Event Table
- SQ Detector 2
- Empower 3, SR1, FR1, SR2HF1, SR2, SR2HF1
- SQD2 ICS v1.51.1644
- Xevo TQ-S micro
- MassLynx SCN945
The time for last event is in the event table is the same as the end time for the Empower instrument method or MassLynx experiment setup.
FIX or WORKAROUND
Change the event time for the last event to be at least 0.01 minutes less that the end time of the Empower instrument method or MassLynx experiment setup.