- “Invalid noise spectrum” error in the message center when processing data
- In a report method or in review, information regarding purity, such as purity angle and threshold, is missing
- Processing code error P01 ("Problem calculating PDA noise vector, cannot continue PDA processing") in the review window
In the noise spectrum, at least one wavelength has a zero value.
FIX or WORKAROUND
Change the data rate. (A data rate of 1 is not recommended.)