- Events tab no longer allows for wavelength or channel entries. See the following images.
Events as seen using v1.50
Events as seen using v1.60 & v1.65
- ACQUITY PDA
- Driver Pack 4, v1.60
- Driver Pack 4 Service Release 1, v1.65
- Empower 3 Enterprise
FIX or WORKAROUND
- Upgrade to Driver Pack 2016 Release 2; this defect was corrected in ICS v1.68, which was released in Driver Pack 2016 Release 2.
The work-around is to collect multiple 2D channels or collect 3D data and extract each channel separately.