- Sample set is run
- Channels are brought into Review
- No chromatograms appear
- Project is newly created
- Empower 3
- ACQUITY UPLC with PDA Detector
The Photo Diode Array option is not enabled for this project.
FIX or WORKAROUND
- Go to Configure the System.
- Find the project, select it, and right-click it. Select properties.
- On the General tab, in the lower-left corner for "Enabled Options", select the check box for Photo Diode Array.
- Close any opened project windows.
- Reopen the project and confirm that you can now review the data.