After upgrading to Agilent ICF 3.0, the area and height reported for acquired FLR data in Empower are significantly lower than those reported when using Agilent ICF 2.2.
- Empower 3
- Agilent ICF 3.0
Incorrect scaling factor; opened as defect PCS 56310
FIX or WORKAROUND
Workaround: Increase the number of decimal places that are displayed, and/or create custom fields to multiply the areas and heights by the missing scaling factor.