- Acquisition run time in Empower stops before each injection is complete with 7890B GC and 7697A Headspace
- The sample set proceeds to the next injection with no error messages or failures
- The channels associated with the injections have data
- GC oven time matches the cycle time of the headspace
- Run Time in Sample set matches method
- Empower 3 Feature Release 3
- 7890B GC
- 7697A Headspace
- GC driver 2.6
- Headspace Control Option 3.0
There is a defect for this behavior: PCS 55142.
FIX or WORKAROUND
- None. Resolution fixed in ICF v3.0 driver.