- After upgrading ICS driver for GC 6850 to 1.40, sample set stops after first injection, and cannot make multi-vial injections
- The first movement to V1 works and injects, but movement to V2 fails
- Error message: "GC setup failed, Instrument Failure, Inlet: Invalid Command"
- Agilent GC 6890
- Agilent 6850 Instrument Control Software Version 1.40
See PCS 55181.
FIX or WORKAROUND
Upgrade to firmware version A.06.02 minimum.
For Agilent GC 6890 with RS-232 control, recommend to upgrade to firmware version A.03.07. This has not been confirmed to workaround this issue, but suspect it will.