Error : "Agilent LC Setup has failed, due to discrepancies between method configuration and on-line instrument configuration."
Error: Agilent LC Communication: A Configuration Change Event has been reported. An instrument has been removed or added to the current system. It is recommended that you Re-Identify your system to detect the new configuration."
- Empower 2
- Empower 3
- Agilent LC Control Software v1.06.1007
- A1100 firmware vA.06.04
- System is defined as A1100
FIX or WORKAROUND
Move to instrument control using Agilent Instrument Control Framework (ICF), and define systems as an AgilentLC instead of A1100.
- Minimum ICF is v2.13