When running GC 7890B or 7890A with Post Run selected in Oven tab of processing method, instrument failure occurs at the end of the injection before loading the next injection.
- Agilent 7890B
- Agilent 7890A
Post Run, when selected, causes the sample set to stop.
FIX or WORKAROUND
Deselect Post Run in Oven tab of Instrument Method.
- Work-around: In order for a post run temperature program to take effect, you must add the program to the temperature ramps table and extend the run time appropriately.
The error 40: SubscribeToData was also reported where Post Run is in use.
This issue has also been reported with G1888A and 7890 configuration
Refer to PCS 54590 for 7890B and PCS 54944 for 7890A