When running GC 7890B or 7890A with Post Run selected in Oven tab of processing method, instrument failure occurrs at the end of the injection before loading the next injection.
- Agilent 7890B
- Agilent 7890A
Post Run, when selected, causes the sample set to stop.
FIX or WORKAROUND
Deselect Post Run in Oven tab of Instrument Method.
- Work-around: In order for a post run temperature program to take effect, you must add the program to the temperature ramps table and extend the run time appropriately.
The error 40: SubscribeToData was also reported where Post Run is in use.
This issue has also been reported with G1888A and 7890 configuration.