- The Beckman CE appears to communicate with Empower
- When the user starts the injection sequence, Empower never leaves "Waiting for Injection" status
- Empower 3
The instrument method is not configured correctly.
FIX or WORKAROUND
In the instrument method, on the Time Program tab, the last Event line in the table should be "End". Without the end line, the instrument will not inject.