- Gas leaks from probe adjuster assembly
- Collar that holds probe to adjuster assembly is loose
- Source pressure test failed error message
- Xevo TQD
Collar of probe is not secured to the probe adjuster assembly.
FIX or WORKAROUND
- Rotating it fully clockwise, tighten the probe collar to secure it to the adjuster assembly.
- Clear the error in the ACQUITY MS console.
- Open and close the source door.