- IntelliStart failed Resolution setup
- Resolution failed for peak at m/z 772.4
- Peak width at half height for:
- m/z 172 fwhh = 0.77
- m/z 772 fwhh = 0.54
- m/z 1072 fwhh = 0.53
- m/z 1971 fwhh = 0.58
- Found: peak intensity at m/z 172 = 1.41 e8
- Found: peak intensity at m/z 772 = 1.29 e8
- Xevo TQ-S
- IntelliStart Resolution Setup
The spectral peaks at m/z 172 and 772 are saturated.
FIX or WORKAROUND
- Adjust setting on Resolution tab in tune page to bring peaks within specification.
- Adjust probe position to decrease the signal strength.
For assistance, contact the local Waters Technical Service group.