- Software says system is vented, but the pumps are still running
- Synapt G2-Si with Edwards XDS35i and XDS100B scroll pumps
- Xevo TQ-S
The covered on/off switch on the base is set to "I" (always on).
FIX or WORKAROUND
- Switch to the other position (software control).