OBJECTIVE or GOAL
Define the transmission window for MSMS acquisition
- Time of Flight mass spectrometers
- On Tune page, look for HM resolution or LM resloution.
- Setting of 1 translates to approximately a 15 Dalton window around the set mass.
- Setting of 5 translates to approximately a 5 Dalton window around the set mass.
- Setting of 15 translates to approximately a 1 Dalton window around the set mass.