- Error message “Degasser no vaccum” appears when powered on.
- Degasser pressure display does not change
- Resetting the instrument does not resolve
- ACQUITY UPLC BSM
- ACQUITY UPLC QSM
Degasser pump (PN: 700001352) failure
FIX or WORKAROUND
- Replace the degasser pump (PN: 700001352) and repair
Submit a request for repair to your nearest Waters.