This article is a portion of a Troubleshooting Path. To begin troubleshooting this issue, please select the link below.
- Sample Fluidics High Pressure Limit (pressure value)
- Flush path with fittings 2 and 3 loose, prime syringe, but error is still present
- ACQUITY H-Class FTN
- ACQUITY I-Class FTN
Error occurs when the pressure exceeds 210 psi in the low-pressure fluidics path.
FIX or WORKAROUND
Replace needle and needle seal.