- "Sample transducer range (4.0)"
- 8800 psi
- Alliance Separations Module 2695
Loose sample transducer connection or bad transducer cable
FIX or WORKAROUND
Replace sample transducer cable - 2690 PRES TRANSDUCER CABLE, part number WAT270906.
With the transducer disconnected, a "sample transducer range (4.0)" error occurs during initialization. In Diagnostics, the pressure is 8800 psi; in Service Utilities, the transducer voltage is 0 volts.