- Unable to set the flow rate on the 1515 in the Run Samples Window
- In the properties of the Node, all instruments except the 1515 respond "Yes" to "OK?"
- Replacing cable does not resolve the issue
- Cannot ping the IP address of the 1515, but all other instruments get a reply
- Changing the port on the switch does not resolve the issue
Communication glitch in the Ethernet switch.
FIX or WORKAROUND
- Unplug the power cable to the Ethernet switch.
- Power-off the systems.
- Power-off the computer.
- Power-on the computer.
- Log into Windows.
- Power-on each instrument, waiting in between until each component completes the start-up diagnostics.
- Plug the power cable back into the outlet for the Ethernet switch.
- Log into Empower.
- Verify the properties of the nodes, particularly the "OK?" status.