- Pressure trace completely drops out to a flat line at about 0 psi during pump operation
- ACQUITY UPLC BSM
FIX or WORKAROUND
Replace the faulty transducer.
Ensure that there is not a mix of 15kpsi and 20kpsi transducers on the system; this situation can cause sync errors.