- Error: Z-axis not calibrated
- ACQUITY UPLC
The sample manager battery power on CPU is low.
FIX or WORKAROUND
- If the error occurs again after calibration, replace the battery.
- The silver coin 3V battery: part number WAT080443
- The yellow square 3V battery: part number 700001834
- Use the dip extractor tool to remove this, part number WAT010339
2. Calibrate the z axis
Check the serial number of the instrument in console and if it has lost the serial number , enter the serial number after replacing battery .