- Static leak test fails
- Parts from the PM kit are not correcting the failure
- Seals are pressurized
- Alliance 2695
The tubing/fittings from the GPV to the primary pump head are faulty.
FIX or WORKAROUND
- Change the defective tubing/fittings.
- Perform the leak test again.
The part number for the 2695 Standard Flow Gradient Proportioning Valve inlet tubing is WAT270986.