- 2707 Autosampler hangs in "Waiting for injection" status
- Empower 3
- 2707 Autosampler
Mismatch between 2707 ICS 1.30 and an older firmware version
FIX or WORKAROUND
- Run Autoloader to update the firmware.
ICS 1.30.132 would not start injection sequence with FW 1.13. After FW upgrade to 1.15 the instrument worked.