- After loading method, the 515 pump flow rate stays at 0 mL/min
- 515 pump
- Pump control module
- System Fluidics Organizer
Communication failure; the 515 pump is not set to remote mode
FIX or WORKAROUND
- Close MassLynx and turn off the computer.
- Turn off the pump control module or SFO.
- Set the 515 to remote/run mode.
- Turn on the pump control module or SFO.
- Turn on the PC and log in to MassLynx.
- Reset communication.
- Load the method and verify that the 515 starts and stops through MassLynx.