- MassLynx presents the following message in the bottom-right of the sample list window: "waiting for injection start"
- The acquisition does not start
The default software settings must be changed to accommodate Autopurification.
FIX or WORKAROUND
Amend the triggering settings in the inlet editor:
- Open the inlet editor.
- Click on Tools.
- Click on Instrument Configuration.
- Click on Events and Triggering.
- Set the pump to Trigger by Software (see screenshot below).