- MassLynx presents the following message in the bottom-right of the sample list window: "waiting for injection start"
- The acquisition does not start
- Error in Status Log in Collection Control "[---] [---] - Timed out waiting for raw data, check the data file"
The default software settings must be changed to accommodate Autopurification.
FIX or WORKAROUND
Amend the triggering settings in the inlet editor:
- Open the inlet editor.
- Click on Tools.
- Click on Instrument Configuration.
- Click on Events and Triggering.
- Set the pump to Trigger by Software (see screenshot below).
MLYNX, MLYNXV41, SUPMM