- Error: "Needle Seal Move Limit Exceeded"
- Characterizing needle seal brings a "force limit exceed" error
- ACQUITY UPLC H-Class
- UP SM-FTN
- ACQUITY UPLC H-Class Bio
- ACQUITY UPLC I-Class
Needle is improperly routed in needle carriage.
FIX or WORKAROUND
- Reseat the needle in the needle carriage, per the needle replacement procedure found in the FTN Overview and Maintenance Guide.
- Calibrate the needle z axis.
- Characterize the needle seal.