- Sample set hangs up after the equilibrate line for ACQUITY UPC2 system
- The system equilibrates but does not continue to the subsequent injection
- ACQUITY UPC2 System
- Empower 3 Feature Release 4
- Driver Pack 2017
- Column Manager 30S
FIX or WORKAROUND
Add an equilibrate line to the sample set before each column switch. Set the run time to 10 seconds and the flow to 0.